LABVIEW GPIB-USB-HS DRIVER
You have just confirmed communication with your instrument. If there is a problem, you no longer see your device. It is important to choose an alias that you can immediately identify with the intended instrument. Configuration in Windows for USB. MAX makes GPIB instrument detection and control easy by providing tools that help you search for connecting instruments, and send and receive communication with your device.
|Date Added:||17 June 2018|
|File Size:||36.60 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Instruments typically respond with the manufacturer’s name, model name, and various alphanumeric characters that the manufacturer uses to track firmware revisions.
GPIB-USB-HS+ – National Instruments
If your system involves different ground potentials, the voltage difference could surge through the GPIB hardware and cause damage. If it is not, then you know to check cables, confirm power, and verify that gpib-uzb-hs instrument is working properly. When VISAIC initially runs, it automatically finds all of the available resources in the system and lists the instrument descriptors for each resource under the appropriate resource type.
Solved: GPIB-USB-HS windows 7 driver ?? – Discussion Forums – National Instruments
Configuration in Windows for USB. If you have two or more instruments on the bus, you can disconnect all instruments except one to determine its address.
The best way to achieve this is by using instrument drivers. These productivity tools help you focus on developing your test applications instead of tracking down obscure problems or establishing communications with your instruments.
Instruments must be powered on and connected to the GPIB controller to be recognized.
Alternatively, system power may be disrupted for a variety of reasons or the actual instrument may begin to malfunction. MAX makes GPIB instrument detection and control easy by providing gpb-usb-hs that help you search for connecting instruments, and send and receive communication with your device.
ID querying is discussed in the next section.
Your instrument should return an identification string and a return count. Unfortunately, engineers sometimes forget to reconnect all the cables and power up all the instruments. Because the GPIB signals at each end are transformed into fiber-optic signals, each unit can reside at a different ground potential.
This indicates that you should check for loose or disconnected cables, power disruption, or a malfunctioning instrument. If there is a problem, you no longer see your device. The basic functions and concepts described above may seem simple; however, they can be invaluable in troubleshooting and getting your GPIB system up and running.
If your computer is already running, the operating system automatically detects the GPIB interface.
To prevent damage to the GPIB-USB hardware or other components in your system when different ground potentials are involved, do any of the following:. Once you have completed these steps, you need a fast way to progress from interactive mode to programming mode so you can immediately begin writing your tests without a tedious transition process.
To communicate with our instrument at Address 4, follow these instructions:. Once gpib-isb-hs have determined the GPIB address of your instruments, you can easily establish communications to verify that you can send and receive gpib-usb-hx to and from the instrument.
Otherwise, the GPIB interface is detected when you start your computer. Install Instrument Drivers Next. This is especially important for large systems featuring many instruments.